b. Install bearing assembly and box (6).
Reassembly procedures will be in accordance
c. Install board and switch assembly (1).
with standard practices for replacement of elec-
The board and switch assembly (1) in-
cludes harness assembly (4) and switch
Reference should be made to the wiring
d. Install spacers (2) and secure with screws.
4-14. Power Supply Assembly.
a. Assemble cover and component assembly
f. Install BATTERY VOLTAGE and CELL
and secure with screw.
BALANCE switches (3 and 1, figs. 1-2 or 1-3).
t h a t were discon-
b. Connect any wires
c . Install power supply assembly into case
4-19. Indicator Assembly.
and secure with screws on bottom of carrying
Connect cable assemblies (19 and 21, fig.
b. Place indicator assembly in kit and secure
a. Install milliammeter in cover assembly and
with roundhead screws on bottom of case assem-
secure with screws and nuts.
b. Connect wires to milliammeter.
4-16. Oscillator-Amplifier Assembly.
or 4-15) and secure with screws at lower end of
assembly with flathead screws.
After all cables are connected, plug kit
into appicable power source and press
a. Place into position in bottom of cover as-
hand down on cell while checking nee-
sembly and solder three leads as marked TOP
dle on milliammeter scale. A slight posi-
TERMINAL, MID TERMINAL, and BOTTOM TER-
tive movement of needle should be indi-
cated when cell is pressed.
b. Press down and secure on panel with
Refer to TB 43-180.
a. Install disc, dial, drive assembly (3, fig. 4-