Disconnect the bypass hose from the side of the sampler and
unscrew the sampler base. Hold the sampler in an upright position and
remove the monitor.
Figure 2.14 Removing the Monitor
2.3.9 Pumping the Monitor Dry
Securely fit the bottom or spoke-side of the monitor onto the
syringe valve (Figure 2.15) and pump the syringe arm point the syringe
away from you and pump the arm, holding the extended arm for 5 seconds
before releasing each time. Pump the arm two or three times, or until the
filter inside the monitor appears dry of fuel.
Drain the excess fuel into a proper drain or receptacle.
Remove the monitor from the syringe.
2 - 1 3