4-29. REMOVE/INSTALL REID VAPOR PRESSURE BATH.
This task covers:
General Mechanics Tool Kit, Appendix B, Item 1
ECU Damper Assembly Removed (paragraph 4-17)
General Safety Instructions
Before performing any maintenance actions on electrical equipment
ensure all electrical power has been turned off. Death or serious
injury may occur from failure to do this.
Remove Reid Vapor Pressure (RVP) Bath. See figure 4-22.
Open screen door to right of cabinet No. 8 and unplug power cord from wall receptacle.
Open screen door to left of cabinet No. 8. Loosen hose clamp on RVP bath drain hose.
Disconnect RVP bath drain hose.
Remove two mounting screws and washers.
Two or more people are required to remove RVP bath because of its
weight. Failure to do so may result in serious injury to personnel or
damage to equipment.
Lift RVP bath up from cabinet.
Remove drain fittings.
Install RVP Bath. See figure 4-22.
Serious injury to personnel or damage to equipment may occur unless
two or more people are used to replace RVP bath because of weight of
Install drain fittings.